- Short report
- Open Access
Safety and efficacy of cervical 10 kHz spinal cord stimulation in chronic refractory primary headaches: a retrospective case series
© The Author(s). 2016
- Received: 17 May 2016
- Accepted: 30 June 2016
- Published: 8 July 2016
Paresthesia-free cervical 10 kHz spinal cord stimulation (HF10 SCS) may constitute a novel treatment modality for headache disorders, when pharmacological approaches fail. We report the results of a retrospective analysis assessing the long-term safety, tolerability and efficacy of HF10 SCS in a group of patients with chronic refractory primary headache disorders.
Four patients with chronic migraine (CM), two with chronic SUNA (Short-lasting Unilateral Neuralgiform headache attacks with Autonomic symptoms) and one with chronic cluster headache (CCH) refractory to medical treatments, were implanted with cervical HF10 SCS. Pre- and post-implantation data were collected from the medical notes and from headache charts. At an average follow-up of 28 months (range: 12–42 months) we observed an improvement of at least 50 % in headache frequency and/or intensity in all CM patients. One SUNA patient became pain free and the other reported at least 50 % improvement in attacks frequency an duration. The CCH patient reported a significant reduction in CH attacks duration. Two patients underwent a surgical revision due to lead migration.
Paresthesia-free high cervical HF10 SCS appears to be a long-term safe and likely effective therapeutic approach for patients with chronic refractory primary headache disorders. These results warrant further prospective studies in larger series of patients.
- Chronic migraine
- Cluster headache
- SUNA syndrome
- Spinal cord stimulation
- Refractory headaches
Chronic daily headache (CDH) is an umbrella term that refers to headache disorders that occur on 15 or more days per month for more than 3 months. Primary CDH is a major worldwide health problem that affects 3–5 % of the population . Chronic migraine (CM) is the most common and disabling primary CDH [2, 3]. Less frequent, but possibly more disabling primary CDH conditions include the chronic forms of trigeminal autonomic cephalalgias (TACs). This group encompasses cluster headache (CH), short-lasting unilateral neuralgiform headache attacks with conjunctival injection and tearing (SUNCT), short-lasting unilateral neuralgiform headache attacks with cranial autonomic features (SUNA), paroxysmal hemicrania (PH) and hemicrania continua (HC) .
Despite advances in the management of headache disorders a significant minority of patients fails to respond or tolerate conventional medical approaches, falling into the category of medically “refractory” headaches. The definition of refractory headaches is an ongoing matter of debate and numerous attempts to draw a consensus have been proposed [5, 6]. For this group of patients, neurostimulation therapies targeting peripheral or central nervous system structures have been emerging as potential treatments . Occipital nerve stimulation (ONS) has been used as first line neurostimulation treatment for the management of various primary headache disorders, including CM, chronic CH (CCH), HC, SUNCT and SUNA [8, 9], based on the encouraging experience of open-label studies. However three large randomized controlled trials (RCTs) testing the efficacy of ONS for CM prevention, displayed modest efficacy, although the studies were criticized for poor methodological designs [10–12].
High frequency (10Khz) spinal cord stimulation (HF10 SCS) is a paresthesia-free neurostimulation therapy, which has been shown to be effective in some chronic pain conditions [13–15]. In view of the mixed outcome of ONS, we began offering high cervical HF10 SCS as part of our neurostimulation approaches to patients with medically refractory chronic primary headache syndromes and participated in a prospective exploratory study testing cervical HF10 SCS in 17 refractory CM. The study showed promising results in terms of tolerability and effectiveness of the therapy . We report here the long-term follow-up of the patients treated in our headache centre with cervical HF10 SCS.
Seven patients with primary CDH were considered suitable candidates for neurostimulation and were offered a trial of HF10 SCS therapy. All patients were diagnosed according to the International Headache Society (IHS) classification criteria  by two, experienced, headache neurologists (SC and GL). Four had CM, two patients had chronic SUNA and one patient had CCH. Patients were considered refractory to conventional pharmacological treatments, according to the proposed definition at the time of screening . Botulinum toxin type A (Botox®) was also tried according to the PREEMPT paradigm  in three out of four CM patients, before considering them refractory . All patients had highly disabling headaches according to the Headache Impact Test (HIT-6). As part of our neurostimulation pathway, patients clinically suitable for invasive neurostimulation attend a cognitive behavioural therapy based implant preparation programme . Six out of seven patients completed the programme before undergoing the stimulation trial. This study was an audit of outcome and, as such under UK guidelines, did not require ethics committee approval.
Subjects with a successful trial were subsequently implanted with an internal pulse generator. The proximal end of the tunnelled lead(s) were connected to new sterile extensions which provided connection to the implantable pulse generator (IPG) (Senza™ system, Nevro Corp., Menlo Park, USA), which was implanted subcutaneously either in the anterior abdominal wall or the gluteal region.
Data collection and follow-up
Baseline headache characteristics, including headache frequency, severity, duration and medications consumption, were collected retrospectively from patients’ records, headache charts, outpatient visits and telephone clinics. Baseline headache-related disability scores were gathered from HIT-6 questionnaires. After the operation, patients were asked to fill a headache diary. More recently CM- and a TACs-specific diaries were designed in our headache centre, hence a daily headache diary and an attack diary were respectively given to CM and TACs patients. HIT-6 questionnaires were administered to patients before scheduled follow-ups. The outcomes used to assess the effectiveness of HF10 SCS for the migraine patients include: change in number of headache days and migraine days, defined according to International guidelines  together with change in headache severity and reduction in abortive treatment consumption. For the TACs patients we assessed change in number of daily attacks, change in duration and severity of the attacks and change in medication use. Adverse events were collected from patients’ notes.
Clinical characteristics of chronic migraine patients pre- and post-high frequency cervical spinal cord stimulation treatment
Duration of chronic headache at time of implant
Number of preventive treatments failed
Follow-up after surgery (months)
Number of headache days/month
Number of migraine days/month
Average headache severity
Analgesia consumption (days/month)
7 (including ONS)
8–12 (PCM, COD)
2–3 (PCM, COD)
9 (including Botox®)
15 (COD, IBP, TRA)
4 (TRA, ALM)
8 (including Botox®)
Daily (PCM, COD)
6–7 (PCM, COD)
6 (Not willing to have Botox®)
Daily (SUM injections)
1 (SUM injection)
At an average follow-up of 25.3 months (range: 12–40 months) post-implant, all patients reported remarkable benefits, with at least a 50 % reduction in monthly number of headache days and migraine days, which led them to revert to an episodic headache pattern (<15 headache days/month). This was associated with a marked reduction in number of days where patients needed to take abortive medications. Patient 3 was also able to discontinue his preventive treatments few months after device activation. At baseline, all patients were classified as severely disabled according to their HIT-6 scores. At the latest available post-operative follow-up, the HIT-6 scores were reduced in all patients; the reduction was meaningful in two patients that rated their headache disability as of “little and moderate impact” (patient 2 and 4). All patients rated their overall percentage of headache improvement between 50–100 %. Most patients reported headache improvement a few days after the HF10 SCS trial was commenced (patients 1, 3 and 4), whereas in patient 2 different stimulation programmes were tried before achieving the maximum benefit.
Continuous stimulation was used in all cases. Unintentionally, patients 1 and 2 switched off the device on few occasions and experienced an exacerbation of their headache that settled within approximately 1 to 2 weeks after the device was switched on again. Stimulation programmes (10 kHz; 30 μs; 1.5–4 mA) were provided to target the dorsal columns in the area corresponding to C2–C3 x-ray vertebral body. Patients 1 and 3 had one lead implanted and benefited from activation of the top two contacts. Patients 2 and 4 had two leads implanted: one at the midline and one more laterally towards the side mostly affected by the headache: they both benefited from alternating between the two stimulation programmes, one using the top two contacts of one lead and the other using the top two contacts in both leads. None of the patients reported any stimulation-induced sensation while the device was on. There were no reports of serious device-related adverse events. Patient 1 in June 2013 experienced a worsening of his headache: high impedances on all contacts suggested system malfunction, which was caused by lead breakage. A surgical lead replacement led to an improvement of the headache within 1 week of device reactivation. Patient 3 and 4 initially reported mild to moderate discomfort at the IPG pocket, which settled respectively after 3 and 4 months.
Clinical characteristics, medical treatments and MRI outcome in Trigeminal autonomic cephalalgias patients treated with high frequency cervical spinal cord stimulation treatment
Side of pain
Site of pain
Attack frequency (daily)
Pain severity (VRS)
Ipsilateral autonomic features
- Facial oedema
- Facial redness
- Blocked nose
- Facial oedema
- Facial redness
- Facial sweating
Effect of indometacin
- Sumatriptan sc
- SPG block
- IV lidocaine
The second patient was a 43-year old woman with a 12-year history of a headache condition that fulfils the IHS criteria for chronic migraine. Seven years ago a new headache began. The condition was initially occurring with an episodic pattern though after 6 months from the onset, it became chronic, without remission periods. The headache fulfilled the IHS criteria of chronic right-sided SUNA syndrome (patient 2, Table 2). In July 2012 she underwent a successful trial and subsequently a full implant of HF10 SCS (10 kHz; 30 μs; 0.4–2 mA). During the first 16 months of treatment, she reported only two SUNA attacks/month lasting 3–5 min, moderate in intensity (VRS 6/10). The attacks subsided once the stimulator was reprogrammed again. Between month 16 and month 37, two reprogramming sessions were required to maintain complete headache relief. After 42 months, she continues to report almost complete resolution of both the SUNA attacks and the chronic migraine, with no adverse events.
One patient with CH was treated with HF10 SCS. He was a 41-year-old man with a 7-year history of left-sided CCH according to the IHS criteria, which became refractory to medical treatments (patient 3, Table 2). He was offered a trial with occipital nerve stimulation (ONS) in March 2011. He became headache-free for 6 months following the implant before the CH relapsed. Despite re-programming, the ONS failed to improve his condition and it was discontinued. In view of the severity of his condition, he was trialled with HF10 SCS (10 kHz; 30 μs; 0.5–1 mA). This was successful, leading to full implant. He was then headache free for 9 months. However the CH attacks gradually came back, occurring at the same frequency and severity of the baseline, but with a significant reduction in duration: from 40–180 min at baseline, to 20–35 min with active HF10 SCS. He rated the improvement at 50 %. Since the implant, trials of sodium valproate and levetiracetam were implemented to improve the condition further. Unfortunately those were not tolerate even at low doses and therefore discontinued. The device was switched off in one occasion for 1 month, leading to a worsening of the CH. A lead revision was necessary after a lead migration occurred 11 months after the surgery.
The management of patients with primary headache conditions refractory to pharmacological treatments remains challenging. ONS is currently considered the surgical treatment of choice for refractory CM and chronic TACs . However its acceptance has been limited by the outcome of methodologically weak RCTs. It is virtually impossible to design good, blinded controlled studies for ONS because the therapy produces paresthesia. A concern with ONS therapy is complication rates. Long-term follow-up of the largest ONS clinical trial conducted in CM showed a total of 183 device-/procedure related adverse events during the study, with 29 lead migrations. Other hardware-related complications include IPG migration, lead breakages or fractures, lead or extension disconnection or malfunction, programmer malfunction or IPG malfunction. Eighty-five adverse events (40.7 %) required surgical intervention and 18 adverse events (8.6 %) required hospitalization . The high rate of hardware-related complications and surgical revisions has recently led the European Notified Body to remove intractable CM as an indication for ONS therapy .
HF10 SCS is a paraesthesia-free system, which has been shown to inhibit evoked afferent nociceptive inputs by modulating wide-dynamic range neuronal activity in the spinal cord of different animal models . HF10 SCS has shown remarkable clinical efficacy in human refractory back pain [13, 15]. When applied to the cervical epidural space, it has been also reported as safe and effective in a series of subjects with upper limb neuropathic pain . We postulate that stimulation directly to the dorsal columns at the level of C2-C3 vertebral bodies with HF10 SCS may modulate the trigeminocervical complex and in turn have a therapeutic effect on primary headaches.
This case series provides initial evidence of effectiveness of HF10 SCS in the management of refractory primary headache disorders. In the treated CM group we observed a significant reduction in headache and migraine days, abortive treatment intake and headache-related disability. The four CM patients all reverted from chronic headache pattern to episodic. Arcioni et al reported promising results using HF10 SCS in a small group of CM patients with medication overuse. Fifty percent of patients reported at least a 30 % reduction in headache days at 6 months. Furthermore HF10 SCS therapy led to reduction of analgesia consumption and disability headache-related in responders . Taken together, these data indicate cervical HF10 SCS as a new, promising treatment in refractory CM. A caveat of the outcome analysis for both studies could be the contribution of the medication withdrawal effect. However in our study, three out of four patients had undergone pre-implantation medications withdrawal without any noticeable change in the headache pattern, suggesting that the improvement is more likely related to HF10 SCS, rather than to the reduction in abortive medications intake.
To date there are no published data on effectiveness of HF10 SCS in TACs. However, high cervical low frequency SCS has shown promising outcome in a small series of seven refractory CCH patients . At a median follow-up of 12 months, the Authors observed a meaningful improvement of the headache in all the implanted patients, allowing reduction in medications intake. Three chronic refractory TACs patients were treated with HF10 SCS in this case series. We observed long-lasting and meaningful benefits in two SUNA patients. The CCH patient, after an initial pain-free period of a few months, experienced reduction in the duration but not in the intensity and frequency of the attacks as the final outcome. The CCH case had been refractory to many other treatments including ONS, thus the partial improvement achieved was been rated as meaningful by the patient.
HF10 SCS may have potential clinical and technical advantages compared to ONS. The HF10 SCS post-implantation therapeutic effect seems to occur almost immediately. A rapid response was noted also in the HF10 Arcioni et al study  and in the Wolter et al CCH series, using a low frequency stimulation device . In contrast, the ONS effect seems to take several weeks or months to develop [9, 26]. This suggests that a short trial treatment may be predictive of outcome with central neuromodulation techniques, but not with peripheral ONS. However, a very high positive trial rate, like that seen in our series (100 %) as well as in the Arcioni et al study , could obviate the need for a stimulation trial, given the additional risks and inconvenience to patients and costs to the health care system . The faster post-implant response of cervical high- and low-frequency SCS compared to ONS, may reflect the anatomical difference in treatment target, suggesting that direct stimulation of the spinal cord may recruit the trigemino-cervical complex more efficiently.
In this case-series two out of the seven patients (29 %) developed hardware-related complications that required additional surgery. Although based on very small numbers, this complication rate is in line with the one reported by Arcioni et al (24 %) , and is lower than reported in ONS studies (40.7 %) . The epidural, axial placement of HF10-SCS leads could be expected to have a lower complication rate than see with ONS leads which are placed at the mobile cranio- cervical junction.
Patients tolerated the paraesthesia-free stimulation well. Historically, the presence of stimulation-produced paresthesia has been considered essential for achieving analgesia in headache neurostimulation studies . The effectiveness of a paresthesia-free neurostimulation system implies that paresthesia sensation may not be required to achieve analgesia . Since the paraesthesia sensation associated with traditional low-frequency ONS is at times considered uncomfortable and may lead to therapy interruption, using a paresthesia-free device may be better accepted by patients.
Limitations of this case series include its retrospective nature and the absence of a control group, raising the possibility that the effect of HF10 SCS in our patients might be attributable to placebo or natural history. Several observations suggest differently: a protracted chronic phase, lack of response to several other treatments, relatively high response rate, sustained long-term improvement, and the rapid deterioration of the headache after the device was switched off or had technical failures.
In this small case series of medically refractory primary chronic headache subjects, the long-term HF10 SCS therapy was well tolerated and beneficial, with an acceptable safety profile (at least comparable with existing implantable neurostimulation devices). In view of the considerable need for improved treatments for this highly disabled group of patients, further prospective research in this field is warranted.
CCH, chronic daily headache; CDH, chronic daily headache; CM, chronic migraine; HC, hemicrania continua; HF10 SCS, 10 kHz spinal cord stimulation; HIT-6, Headache Impact Test; IHS, International Headache Society; IPG, implantable pulse generator; ONS, occipital nerve stimulation; PH, paroxysmal hemicrania; RCTs, randomized controlled trials; SUNA, Short-lasting Unilateral Neuralgiform headache attacks with Autonomic symptoms; SUNCT, Short-lasting Unilateral Neuralgiform headache attacks with conjunctival injection and tearing; TACs, trigeminal autonomic cephalalgias; VRS, visual rating scale
The authors would like to acknowledge the contribution of Dr Sam Chong (SC in the manuscript) for helping in the diagnosis and management of some of the patients in this study.
This research received no specific grant.
GL: patients diagnoses and follow-ups, data acquisition/interpretation and manuscript drafting/revision. MT: patients follow-ups, data acquisition and interpretation and manuscript drafting/revision. SP: analysis/interpretation of data and manuscript drafting/revision. TS: implanted some of the patients, analysis/interpretation of data and manuscript revision. AAK: implanted some of the patients, analysis/interpretation of data and manuscript revision. All authors read and approved the final manuscript.
Dr Lambru has received speaker honoraria and funding for travel from Allergan, Electrocore and Nevro Corp. Dr Trimboli: nothing to declare. Dr Palmisani received speaker fees and/or sponsorships to attend professional meetings from Nevro Corp and Medtronic. Dr Smith has done consultancy work for Nevro Corp and has been an investigator in Nevro sponsored studies. Dr Al-Kaisy has received travel sponsorship and speaker fees from Medtronic and Nevro Corp, and he is the principal investigator in separate studies sponsored by Medtronic and Nevro Corp.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Scher AI, Stewart WF, Liberman J, Lipton RB (1998) Prevalence of frequent headache in a population sample. Headache 38:497–506View ArticlePubMedGoogle Scholar
- Vos T, Flaxman AD, Naghavi M et al (2012) Years lived with disability (YLDs) for 1160 sequelae of 289 diseases and injuries 1990-2010: a systematic analysis for the Global Burden of Disease Study 2010. Lancet 380:2163–2196View ArticlePubMedGoogle Scholar
- Steiner TJ, Birbeck GL, Jensen RH, Katsarava Z, Stovner LJ, Martelletti P (2015) Headache disorders are third cause of disability worldwide. J Headache Pain 16:58View ArticlePubMedPubMed CentralGoogle Scholar
- Headache Classification Committee of the International Headache Society (IHS) The International Classification of Headache Disorders, 3rd edition (beta version) (2013). Cephalalgia 33:629–808Google Scholar
- Goadsby PJ, Schoenen J, Ferrari MD, Silberstein SD, Dodick D (2006) Towards a definition of intractable headache for use in clinical practice and trials. Cephalalgia 26:1168–1170View ArticlePubMedGoogle Scholar
- Martelletti P, Katsarava Z, Lampl C et al (2014) Refractory chronic migraine: a consensus statement on clinical definition from the European Headache Federation. J Headache Pain 28:15–47Google Scholar
- Lambru G, Matharu MS (2014) Peripheral neurostimulation in primary headaches. Neurol Sci 35(Suppl 1):77–81View ArticlePubMedGoogle Scholar
- Magis D, Schoenen J (2012) Advances and challenges in neurostimulation for headaches. Lancet Neurol 14:708–719View ArticleGoogle Scholar
- Lambru G, Shanahan P, Watkins L, Matharu MS (2014) Occipital nerve stimulation in the treatment of medically intractable SUNCT and SUNA. Pain Physician 17:29–41PubMedGoogle Scholar
- Lipton R, Goadsby PJ, Cady R et al (2009) PRISM study: occipital nerve stimulation for treatment-refractory migraine. Cephalalgia 29(Suppl 1):30Google Scholar
- Saper JR, Dodick DW, Silberstein SD et al (2011) Occipital nerve stimulation for the treatment of intractable chronic migraine headache: ONSTIM feasibility study. Cephalalgia 31:271–285View ArticlePubMedPubMed CentralGoogle Scholar
- Silberstein SD, Dodick DW, Saper J et al (2012) Safety and efficacy of peripheral nerve stimulation of the occipital nerves for the management of chronic migraine: results from a randomized, multicenter, double-blinded, controlled study. Cephalalgia 32:1165–1179View ArticlePubMedGoogle Scholar
- Al-Kaisy A, Van Buyten JP, Smet I et al (2014) Sustained effectiveness of 10 kHz high-frequency spinal cord stimulation for patients with chronic, low back pain: 24-month results of a prospective multicenter study. Pain Med 15:347–354View ArticlePubMedGoogle Scholar
- Al-Kaisy A, Palmisani S, Smith T et al (2015) The use of 10-kilohertz spinal cord stimulation in a cohort of patients with chronic neuropathic limb pain refractory to medical management. Neuromodulation 18:18–23View ArticlePubMedGoogle Scholar
- Van Buyten JP, Al-Kaisy A, Smet I et al (2013) High-frequency spinal cord stimulation for the treatment of chronic back pain patients: results of a prospective multicenter European clinical study. Neuromodulation 16:59–65View ArticlePubMedGoogle Scholar
- Arcioni R, Palmisani S, Mercieri M et al (2016) Cervical 10 kHz spinal cord stimulation in the management of chronic, medically refractory migraine: A prospective, open-label, exploratory study. Eur J Pain 20:70–78View ArticlePubMedGoogle Scholar
- Headache Classification Subcommittee of the International Headache Society. The International Classification of Headache Disorders: 2nd edition (2004). Cephalalgia 24(Suppl. 1): 9–160.Google Scholar
- Aurora SK, Winner P, Freeman MC et al (2011) OnabotulinumtoxinA for treatment of chronic migraine: pooled analyses of the 56-week PREEMPT clinical program. Headache 51:1358–1373View ArticlePubMedGoogle Scholar
- Palmisani S, Al-Kaisy A, Arcioni R et al (2013) A six year retrospective review of occipital nerve stimulation practice-controversies and challenges of an emerging technique for treating refractory headache syndromes. J Headache Pain 6:14–67Google Scholar
- Silberstein S, Tfelt-Hansen P, Dodick DW et al (2008) Guidelines for controlled trials of prophylactic treatment of chronic migraine in adults. Cephalalgia 28:484–495View ArticlePubMedGoogle Scholar
- Martelletti P, Jensen RH, Antal A et al (2013) Neuromodulation for chronic headaches: position statement from the European Headache Federation. J Headache Pain 21:14–86Google Scholar
- Dodick DW, Silberstein SD, Reed KL et al (2015) Safety and efficacy of peripheral nerve stimulation of the occipital nerves for the management of chronic migraine: long-term results from a randomized, multicenter, double-blinded, controlled study. Cephalalgia 35:344–358View ArticlePubMedGoogle Scholar
- https://www.swissmedic.ch/recalllists_dl/10612/Vk_20141022_21-e1.pdf. Accessed 26 June 2016.
- Cuellar JM, Alataris K, Walker A et al (2013) Effect of high-frequency alternating current on spinal afferent nociceptive transmission. Neuromodulation 16:318–327View ArticlePubMedGoogle Scholar
- Wolter T, Kiemen A, Kaube H (2011) High cervical spinal cord stimulation for chronic cluster headache. Cephalalgia 31:1170–1180View ArticlePubMedGoogle Scholar
- Burns B, Watkins L, Goadsby PJ (2009) Treatment of intractable chronic cluster headache by occipital nerve stimulation in 14 patients. Neurology 72:341–345View ArticlePubMedGoogle Scholar
- Kapural L, Yu C, Doust MW et al (2015) Novel 10-kHz High-frequency Therapy (HF10 Therapy) Is Superior to Traditional Low-frequency Spinal Cord Stimulation for the Treatment of Chronic Back and Leg Pain: The SENZA-RCT Randomized Controlled Trial. Anesthesiology 123:851–860View ArticlePubMedGoogle Scholar